Product review for AED Defibrillators Automatic External Defibrillators for Home/School/Business
AEDs: 5 Best AEDs in 2019
Philips HeartStart OnSite Routine Maintenance – AED Authority
Difficult airway: difficult to AMBU bag?
Foldable 9"x6" Panel AED Wall Sign – Part Number:SSS 021F
Marina Del Rey Baton Training
An Introduction to the Brayden CPR Manikin
CPR and First Aid Training In Illinois and Iowa
Sudden Cardiac Arrest AED Winner | HeartSine Samaritan
medical product catalog Ver.17D for clinic,hospital and home use
Joint Finance / Personnel Committee 10-30-18
(Automatic External Defibrillator) is a defibrillation device, which is safe, portable and easy to use.
It can be applied in family and public place to give first aid for the sudden death symptom.
It can not only enhances the success rate of defibrillation, but also reduce the injury of defibrillation to heart to the utmost extent.
There is low requirement for the operator to use the device. The defibrillator will guide the operator to carry out defibrillation treatment via voice prompt step by step, give safety alert if the occasion arises.
Output：Biphasic Truncated Exponential
Energy Sequence:150, 150, 200J
Charge Time: 8 sec. to 150J; 10 sec. to 200J (New Battery)
Analysis Time: 9 sec.
The maximum time from the initiation of rhythm analysis to readiness for discharge with a new battery: Less than 30 seconds
The maximum time from the initiation of rhythm analysis to readiness for discharge after 6 shocks: Less than 35 seconds
The maximum time from initially switching power on to readiness for discharge: Less than 25 seconds
Audible Prompts: 22 audible prompts
Visual Prompts: LED prompts
Controls: Two buttons-ON/Off, Shock
Output Energy Accuracy: ±15% into any impedance from 25 to 175Ω
The Maximum Voltage: 1200±50V
Output disabled when the patient impedance is outside limits: 20Ωto 200Ω
Battery Part No. CR123A-4*2 Non-rechargeable 12V DC, 2.8Ah
Capacity: At 150 Joules
Dimensions :303*216*89 mm